|Frequency:||2.00Hz - 50.00kHz|
|Operating Temperature (F):||14 - 122 Degrees|
|Range (AC Current):||10A|
|Range (AC Voltage):||6.00mV - 600V|
|Range (DC Current):||10A|
|Range (DC Voltage):||600.0mV - 600V|
|Safety Category Rating:||CAT III 600V|
Do you have questions about this product?
Get answers from real customers and in-house experts with AnswerBox.
Q:how do I replace the 10 amp fuse in a 2216-20 multi meter?
A:You will need to remove the DMM Stand, then the Battery Door then the Bottom Case. The fuse is located on the Main PCA. Here is a link to show you where the fuse is located: http://documents.milwaukeetool.com/54-07-2260.pdf
Q:how do I replace the 10amp fuse
A:Here is a link to the parts listing. This is to show you where the fuse is located. You will need to remove the DMM Stand, battery door and the bottom case to get to the fuse.